The data collection segment coordinates parameters of the automatic exposure providing the optimal load for X-ray tube to the best image quality.
Maximum number of examinations: 100
Maximum series number: 999 including also two supplementary sets of images per each examination.
The highest efficiency is ensured by the image processor with 4000 MIP equivalent and was planned to process high speed images.
The image viewer segment enables table side image viewing with the help of infrared remote control unit.
Maximus ROTALIX tube with mesh wireMRC 04 07 X-ray tube. Content:
Bolus Chase contrast medium tracking
In case of bloodstream, especially towards the lower organs, difficult to estimate for display of vascular structure. This tool
solves the problem of difficult steps derived from the 'the lack of adjustment' between the blood-stream and the selected program,
as well as the lack of real-time projection. During non-stop collection of real-time data the contrast medium tracking is
provided by a continuous motorized table scanning movement with speed adjusted to the blood-stream.
To make the operator's work easier the contrast medium is tracked either by interactive speed control or by programmed speed control when speed-curves can be selected.
Display in subtractive or non-subtractive mode. Interactive tracking of contrast medium.
The scanning speed as well as the frame rates is program-driven and speed-curves adapted to the blood-stream can be selected from the program.
The rotational angiography is a data collection procedure, which enables 3 dimensional insight into the
vascular structure. The C-arm is rotating while collecting images.
One of the data collection programs of the system selects the procedure and the specific parameters. Operator can choose the initial and final positions of the rotation which are memorized by the two-position ACP.
Rotation speed: 30 degree/sec.
Imaging speed of Integris CV: max. 25 (30) image/sec.
High speed (HS) DICOM image interface version 3
An open Integris system enables the integration of the equipment to various networks and information systems.
HSDII Interface provides the forwarding of image files and exposure/records, heart-beat, set of records by the Image Server function to DICOM based destinations such as Radiological (RIS) and cardiologic PACS, Easy Vision 4.2.2 systems or to optional Integris 3D-RA dedicated to Integris V- and Allura.
HSDII enables also storage of DICOM images, DICOM based images and attached information.
Desktop display operating console
Syncra Rouch desktop display operating console with data management for numerous diseases. Display workstation including alphanumerical keyboard to handle patients' data, panel and XB 17 inch monitor, TG17FM monitor serving for displaying and review of high-resolution images.Other advantages:
Angiomat Illumena digital injection system (with stand)
Aim of the Angiomat Illumena injector is to inject a contrast medium impermeable for X-ray into the
vascular system for radiological vascular test or CT examination. According to its configuration the
injector can be used in angio or CT mode. The injector is equipped with flow rate, volume and pressure
The applied injector system can be selected: either 150 ml disposable or 150 ml reusable injectors.
(Type: KODAK DryView 8100, with built-in PrintServer PL 25)